Objetivo: ISO specifies the properties and performance of sterile single-use hypodermic syringes with or without needle, made of plastic materials . Find the most up-to-date version of ISO at Engineering Sterile hypodermic syringes for single use – Part 3: Auto-disable syringes for fixed -dose immunization (ISO ); German version EN ISO
|Published (Last):||22 May 2010|
|PDF File Size:||9.16 Mb|
|ePub File Size:||9.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
Stop the plunger travel and readjust the recorder to zero. IS Part 3: It is designed to cover every economic sector and virtually every activity of the humankind where technical standards may be used.
It is recognized that syringes designed to reduce the risk of needlestick injuries, in addition to preventing sharps injuries, may also comply with this part of standard with regard to their auto-disable properties, but it is stressed that anti-needlestick properties of syringes are not in themselves addressed in this standard.
Summary This part of ISO specifies the properties and performance of sterile single-use hypodermic syringes withor without needle, made of plastic materials and stainless steel and intended for the aspiration of vaccines orfor the injection of vaccines immediately after filling. Attention is particularly drawn to the following: This part of ISO is not applicable to isso for use with insulin specified in ISOsyringes madeof glass specified in ISOsyringes for use with power-driven syringe pumps specified in 7868-3 ,auto-disable syringes for variable dose delivery and syringes designed to be prefilled.
Standards are also reviewed periodically; a standard alongwith amendments is reaffirmed when such review indicates that no changes are needed; if the review indicates uso changes are needed, it is taken up for revision.
The number of significant places retained in the rounded off value should be the same as that of the specified value in this standard.
International Classification for Standards ICS is an international classification system for technical standards.
Syringes with re-use prevention feature This document has been re-assessed by 7886-33 committee, and judged to still be up to date. The timing of the activation of the auto-disable feature may vary by design, typically within the ranges described below: Both the needle and the syringe shall be rendered incapable of re-use after delivery of uso intended fixed dose, under normal conditions of use. For syringes with integrated needle, 14,2 of ISO You will be redirected to our e-shop where you can buy the document.
The corresponding Indian Standards which are to be substituted in their places are listed below along with their degree of equivalence for the editions indicated: Buy Do you want to buy this document?
Water for analytical laboratory use — Specification and test methods ISO Users of Indian Standards should ascertain that they are in possession of the latest amendments or edition by referring to the latest issue of ‘BIS Catalogue’ and ‘Standards; Monthly Additions’.
NBN EN ISO | NBN
This standard is intended to cover ‘fixed dose’ isso syringes that are rendered inoperable after delivery of the intended dose. Sterile hypodermic needles for single use ISO Re-use of injection equipment in the absence of sterilization has increasingly let to transmission of blood-borne pathogens this standard.
Upon delivering a fixed dose of vaccine, the syringe is automatically rendered unusable. Syringes for manual use IS No part of the these publications may be reproduced in ios form without the prior permission in writing of BIS. Connect the needle of the syringe to the rubber stopper of the tubing of the reservoir B.
When a syringe with integrated needle is tested in accordance with Annex 8 of ISO Certain conventions are, however, not identical to those used in Indian Standards.
Syringes for manual use ISO It is important to us that you purchase the right document.
No secondary or additional action on the part of the user shall be required. For the purpose of deciding whether a particular requirements of this standard is complied with, the final value, observed or calculated expressing the result of a test or analysis, shall be rounded off io accordance with IS 2: In this adopted standard, reference appears to certain International Standards for which Indian Standards 786-3 exist. Adjust the relative positions of the syringe and reservoir so that the water level in the reservoir is approximately level with the mid-point of the syringe 7886-3.
Do you want to buy this document? Other standards can be applicable when syringes are used for any other intended purpose than those specified in ISO In all cases, once the auto-disable feature has been activated: Figure 1 — Schematic representation of auto-disable syringe for fixed dose IS Part 3: